3-Point Calibration for accuracy
The calibration laboratory techniques established for Temprecord ISO17025 accreditation have been developed to make 3 Point Calibration a standard process when manufacturing all General Use Scientific and RH data loggers.
After initial assembly all General Use, Scientific & RH data loggers are calibrated at -15 °C, 0 °C and +40 °C, or -38 °C, 0 °C and +40 °C for low temperature probes with the option of specified points such as -80 °C. This process provides an accuracy of within 0.2 °C (- 0.36 °F) across a range of -20 °C to +50 °C (-4 °F to +122 °F) or for low temperature probes -40 °C to +50 °C, (-40 °F to +122 °F).
From a data logger showing results like this we would re-calibrate the data loggers programming at the three points: -15 °C, 0 °C and +40 °C, or -38 °C, 0 °C and +40 °C for low temperature probes.
As can be seen, this process straightens out the accuracy curve dramatically giving the end-user an accurate product across the stated range.
This level of accuracy in the data logger can only be achieved when the data logger’s software has the ability to have these correction factors applied to it.
Currently Temprecord is one of the few manufactures in the world that applies the correction data to the individual data logger economically.
Some manufacturers do provide calibration certificates but these certificates only provide the correction factors at any nominated points. They cannot make allowances for the correction.
Without this, a calibration certificate is only telling you how inaccurate your data logger really is.